Most meters are in common ground design. The volt meter is able to monitor it's own supply voltage simply by connecting the "+" input to the power (+) terminal and leave the Negative (-) unconnected.
For an amp(current) meter, you got to be careful about the looping feedback from the shunt. To avoid the problem, you better to install a dc-dc isolated power module to provide it an independency power to the meter
1 post • Page 1 of 1
- Panel Meter connection.pdf
- (30.09 KiB) Downloaded 2152 times
Who is online
Users browsing this forum: No registered users and 1 guest